Wink of Knowledge: High Density Media – DLO Density Meter for Liquids
Volume 2 | Number 3
Why this test?
The measurements carried out show that our DLO measures very precisely even in media with a density far above the previously specified maximum value.
What is a wink of knowledge?
Do you know the need to have to sometimes measure, draw or do something quickly? The speed to the result counts more than the perfect (scientific) approach. For this reason, we have introduced a wink of knowledge. Science with a wink, so to speak. We don’t want to prove anything scientifically, but quickly demonstrate something pragmatically. If you are interested, we would be happy to discuss these results in more detail with you and your project.
What liquids were used?
• Tetrachloroethylene (Carl Roth, item no.: 4737.1)
- Carl Roth, item no.: 4737.1
- Molar mass: 165.83 g/mol
- Density: 1.61 g/cm³
The density measurement was carried out with the DLO-M1 density sensors for liquids. For this purpose, the sensors were each flushed with tetrachloroethylene. Using the logging function, one measured value per second was recorded for density and temperature. As a reference, the density was measured with the laboratory measuring instrument DSA 5000 M (Anton Paar). The reference values measured at 20 °C and 30 °C were linearly interpolated to obtain the temperature-dependent density of tetrachloroethylene.
The TrueDyne sensor
The DLO-M1 sensor measures the density of liquids in a microelectromechanical system (MEMS system). Within the MEMS system, the liquid is directed to an omega-shaped microchannel, the so-called omega chip. This tiny silicon tube – it is hardly thicker than a hair – is set into oscillation for the measurement. The density of the medium can be derived from the natural frequency of this oscillation: the denser the medium, the lower the oscillation.
The measuring system in submillimetre size enables the compact construction of the sensor. It measures just 80 x 30 x 15 mm and thus fits into even the tightest of spaces. The measured values reach the higher-level system via an RS232 interface and in the ASCII command protocol in the TrueDyne Sensors standard.
- Reference density measurement with laboratory density meter DSA 5000 M (Anton Paar)
- Inserting the sensor into the measurement setup according to figure 1
- Pumping the tetrachloroethylene through the density sensor by means of syringes
- Syringe with tetrachloroethylene
- Density sensor DLO-M1
- Data evaluation
- Return of the medium
- Syringe takes up the test fluid again
The measurement results are shown in Figure 2. The black dashed line marks the temperature-dependent reference density, which was determined with the laboratory measuring instrument DSA 5000 M (Anton Paar). The solid black lines mark the reference density with a tolerance of ±0.5 kg/m³ (±0.0005 g/cm³). This corresponds to the maximum measurement deviation of the TrueDyne density sensor DLO-M1.
The coloured dots mark the readings of three different TrueDyne DLO-M1 sensors. It should be noted that dynamic measurement deviations occur during the flow through the sensors: Due to the self-heating of the sensor, the sensor temperature deviates from the temperature of the incoming, colder fluid. At lower flow rates, these two temperatures converge so that in the static case the measurement deviations from the reference density values are less than ±0.1 kg/m³ (±0.0001 g/cm³).
The measurement results shown demonstrate that the TrueDyne DLO-M1 sensors achieve the specified accuracy of ±0.5 kg/m³ in density measurement even far beyond the specified density range (>1600 kg/m³ instead of ≤1000 kg/m³). By compensating for the self-heating of the sensor, even accuracies of ±0.1 kg/m³ are possible.
Do you have applications in this extended measuring range? Get in touch with us!
Sensors that might interest you
Applications that might interest you
Other articles that might interest you
Density and CO2 formation were measured during the fermentation process. The measurements of the wort’s density, degree of fermentation and alcohol content agree very well with expectations and verification measurements with laboratory equipment. The presented solution enables automatable, continuous monitoring of the process as well as determination of …
Wink of Knowledge: Measurement of the hydrogen peroxide concentration in water with <0.025% measurement uncertainty
Density measurements of hydrogen peroxide in water with the VLO-M2 were correct within <0.05 kg/m3 using control measurements from a laboratory analyser. The measured deviations correspond to a measurement uncertainty of <0.025% (250ppm) for the concentration determination.
Article: In-line measurements of the physical and thermodynamic properties of single and multicomponent liquids
Microfluidic devices are becoming increasingly important in various fields of pharmacy, flow chemistry and healthcare. In the embedded microchannel, the flow rates, the dynamic viscosity of the transported liquids and the fluid dynamic properties play an important role. Various functional auxiliary components of microfluidic devices such as flow restrictors, valves and flow meters need to be characterised with liquids used in several microfluidic applications.
Flow sensors based on the calorimetric measuring principle typically have to be adjusted to a specific medium. This wink of knowledge shows that a real-time density measurement is highly suitable for correcting the measured flow value for pure gases as well as binary gas mixtures.
Article: Design, Simulation, Fabrication and Characterization of piezoelectric MEMS Cantilever for Gas Density and Viscosity Sensors Applications
Design, Simulation, Fabrication and...
The aim of the study is to develop a compact, robust and maintenance free gas concentration and humidity monitoring system for industrial use in the field of inert process gases. Our multiparameter gas-monitoring system prototype allows the simultaneous measurement of the fluid physical properties (density, viscosity) and water vapor content (at ppm level) under varying process conditions.
MEMS technologyAt a glance In the previous...
The resonator density measurement At a glance In...
Density measurement basics At a glance This...
A rock in the surfRethink Sensing is not only...